Sometimes it is necessary to transport sample from some distance to the analyzer. There is no inherent problem in doing this, unless the sample is subject to condensation (see “Sample Conditioning” article); the analyte is unstable; or the interconnect tubing material might exhibit chemisorption properties.
System lag time, reflecting the time for gas sample to travel through interconnect tubing, from a remote point back to the analyzer, can be estimated as follows, with the familiar πr2h (volume of a cylinder) formula:
100 feet (30.48 m) of typical 0.250 inch OD x 0.125 inch ID tubing (6.4 x 3.2 mm) contains a volume of 0.2415 liters. Assuming a flow rate of 0.5 liter/minute, it will take 0.483 minutes (29 seconds) for the remote sample to traverse the tubing.
Various methods are available to decrease lag time, including sampling at a higher flow rate and bleeding off to the analyzer, and incorporating heavier duty pumps and wider bore tubing.
For further information, please contact the factory.