No analytical method is completely specific. Gases present in the environment, other than the “target” gas of measurement, may affect instrument response. Interferences (aka “cross-sensitivities” are not necessarily linear, and may also exhibit time dependent characteristics.
The charts that follow detail the approximate concentration in parts per million of interfering gas required to cause a 1 ppm deflection in the chosen analyzer. In many cases, specificity can be improved. Please note that the response values given are not absolute, and may vary depending on sensor formulation.
The special case of how alcohols affect electrochemical sensors is discussed in this Knowledge Base article.
For further information on the effects of interfering gases, please contact the factory.
The charts follow the format, and grouping of gases, that was originally established in early Interscan print brochures.
Click on the links below to view each chart.